530 Coating Thickness Gauge
Product description:1.Measurement Range： 01250um or 050mil 2. Accuracy： [(1%~3%)H+1um] Note：H is thickness reading 3. powers down after 60 seconds of no activity. 530 canmeasurethethicknessofnon-magneticcoatinglayerscove
2. Accuracy：± [(1%~3%)H+1um] Note：H is thickness reading
3. powers down after 60 seconds of no activity.
530 can measure the thickness of non-magnetic coating layers covered on magnetic substrate. Such as: non-magnetic (aluminum, chrome, copper, enamel, rubber, paint) covered on magnetic substrate (steel, iron, alloy and magnetic stainless steel)
Keystroke once；Wait until READY！ is displayed on the LCD. Now the gauge is ready for measurements.
Keystroke twice; when BASE ZERO is displayed, place flat on surface; hold steady. After measurement is obtained, DONE is shown and gauge beeps. Then lift gauge.
Note: Please conduct zeroing process after changing base. Zeroing process could be repeated for better results.
3、Two-Point User Calibration
Keystroke three times, enter into the process of tow-point user calibration; when
USER CAL BASE is shown on the LCD, please conduct a measurement on the base first; when
S1 is shown on the LCD, place Shim S1 on the base, conduct a measurement on shim S1 and lift gauge; when
S2 is shown on the LCD, place Shim S2 on the base, conduct a measurement on shim S2 and lift gauge; When DONE is displayed, it is ready for use.
Note: The two-point user calibration process could be repeated for better results.
Keystroke four times, convert from ums to mils or vice versa.
Keystroke five times, gauge resets.